I had a frustrating session this morning with my sensor. I’m wondering if anyone has run into this and has any idea what causes it and how to resolve it.
My pump started alarming about approaching low and low SG at about 2 this morning. Got up and checked my BG a few times over the next hour. BG was fine - in the 115 - 125 range. SG was insisting on 60 - 75. Did not want to try to calibrate it out as auto was going to be a major headache because of the difference in numbers. Took the pump out of auto so I would get some insulin and finally ended up turning off the sensor and going back to sleep. When I woke and reconnected the sensor the SG as in the 40s, BG in the mid-90s. Tried a calibration, had a rejection, did a second one, calibration was accepted (after an “updating sensor”). The sensor finally “sorted” itself out about an hour after getting up. It made for a really long night.
Any idea what causes this sort of sensor behavior. The site area I’m using has provided good sensor results in the past and I do rotate in this area.